Up to 27 pairs of speed and time settings (flow rates) can be programmed in DOSER and up to 99 steps of speed and time can be programmed in HI-DOSER.
The programming mode is accessed by simultaneously pressing the buttons REMOTE and RUN until the indication “PGM” appears on display and the REMOTE and RUN LEDs are switched on.
Remark: If you repeat the following simultaneous pressing of the REMOTE and RUN buttons, the memory will be cleared and the indication “cLE” will appear on the display (Figure 2-2). To enter the programming mode again, press the REMOTE and RUN buttons again until “PGM” appears (Figure 2-1).
To stop the running program definitively, press the RUN button. The RUN and ON/OFF LEDs are switched OFF.
It is possible to stop the pump by pressing the ON/OFF button, to change the rotation speed during any running program step. This allows the reaction in emergency situations.
Remark: Do not forget to switch ON the powder DOSER or HI-DOSER again (by pressing the ON/OFF button), after finishing the intervention.